Seca 103 Instructions For Use Manual
OPERATING CONNECTED SECA MEASURING DEVICES
If you connect seca measuring devices to the seca connect 103, a number of functions of the seca measuring devices may not be used or only used with limitations. Otherwise, the following faults can occur:
Malfunction of individual devices
Transmission of invalid measured values
Incorrect allocation of measured values to patient files
Observe the instructions for use of the seca measuring devices.
Inform your users about these limitations. To do so, use the quick reference “Measurement procedure” ➔ Annex: Quick reference: Measurement procedure.
Performing final work
Once you have completed the retrofitting of the measuring devices, perform the following steps:
Ensure that the device is positioned on the ground flat and stable.
Ensure that no cables or other parts touch the weighing platform.
Perform a function check of the device as described in the respective instructions for use.
Remove all tools and materials.
Clean the floor and other installation areas.
Clean and disinfect the measuring devices as described in the respective instructions for use.
Clean and disinfect your hands.
You have the following options to continue:
Add a seca measuring device in the seca connect 103 software: ➔ Adding a seca measuring device
Change the configuration of a seca measuring device in the seca connect 103 software: ➔ Changing a setting of a seca measuring device
Connecting a seca 452 interface module
Malfunction as a result of faulty multi-functional display
When the device is laid down, the multi-functional display is directly on the floor and may be damaged.
Lay the device down slowly and carefully on a soft surface, a blanket, for example.
1. Lift the measuring station at the rear and carefully tilt it forward.
Malfunction due to installation errors
If cables are strained during installation, e. g. with sharp bends or kinked connectors, this may result in faulty displays and failure of the display.
Route all cables to prevent sharp bends and kinked connectors.
Provide strain relief by routing all cables in the relevant holders.
2. Connect the multi-functional display to the seca